In the process of debugging the strange intermittent display issue on my Tektronix 2246 after re-capping the power supply, things are not looking so good. Things have actually gone from bad to worse. First of all, the short (or open connection) somewhere in this unit which I can return the unit to normal operation by twisting the chassis has been more difficult to find that planned. I pulled and went through the entire power supply again and did not find any issues. Same for all of the connectors, all are seated correctly without issue. Every screw was in place and I checked to make sure there were no poor grounding issues throughout the scope. What happened next is the result of my own stupid mistake (again) which is just making things worse.
In the process of removing the power supply, you have to remove the high voltage anode cable to the crt which is generated in the main power supply. This cable runs out of the supply towards the front of the unit and has an insulated high voltage connector that is clipped on the the metal chassis between the power supply and front panel board. When I was pulling the power supply out for the third time, I wasn't careful and unplugged this connector only a minute or two after powering the scope down. At this point I let the cable go and it fell towards the A16 main processor board letting a nice spark jump from the high voltage cable to the main board most likely due to stored capacitance that had not been bled off yet. I didn't see exactly where it hit, but it didn't matter. The stored voltage had hit the board and I knew damage had been done. Upon reassembly and power up it was confirmed.
The display was in bad shape at this point, everything to the left say 2/5ths of the display was not showing up and squashed into a vertical line. Definitely not good. So now my priorities have changed, now I have to debug this issue to make sure I haven't burnt out something serious like the main display DACs or readout processors as these would most likely be difficult parts to locate and replace.
I started right with the schematic, which was my next stop anyway in locating the intermittent display issue. The full service manual including schematics is easily available for this scope which is another reason why I love the older Tektronix gear. I focused right away at the character generator and display readout circuitry.
After some probing around, I noticed that output from the A16 processor pcb was odd, the horizontal output drive for the display was not looking right, the bottom half of the waveform was clipped which would explain the left half of the display being smashed into a vertical line. Further up the signal path, the signal was looking correct from the DAC and the multiplexer. So the final op amp stage was looking to maybe be the culprit which makes sense as these op amps definitely wouldn't survive a direct high voltage hit.
|Output of mux on left, output after op amp on right|